|pass||% good||op.||total||hardware||software||% recovered||% recoverable||exp||notes|
|9911010649||47.61||Y||52||52||48||Formatter not setting up properly (during tape drive move), no Telemetry housekeeping data from HALCA|
|9911011449||99.36||Y||periods of high niose|
|9911020806||99.06||Y||no Telemetry housekeeping data from HALCA at end of pass,HALCA signal late, periods of high niose|
|9911021619||99.17||Y||periods of high niose|
|9911030923||98.89||Y||1||HALCA signal late, phase glitch, periods of high niose|
|9911041040||99.13||Y||HALCA signal late, phase glitch|
|9911050537||0.00||X||100||100||100||Formatter not accepting MCB commands - hung up command file|
|9911051350||99.39||Y||periods of high niose|
|9911060655||98.94||Y||1||HALCA signal late, periods of high niose|
|9911070812||99.23||Y||HALCA signal late, periods of high niose|
|9911080928||98.69||Y||1||HALCA signal late, periods of high niose, signal lost for 34 seconds at 11h05m22|
|9911091046||99.27||Y||HALCA signal late, phase glitch|
|9911100543||97.10||Y||2||HALCA signal late, periods of high niose, signal lost for 22 seconds at 6h18m07|
|9911110700||98.26||Y||1||HALCA signal late, periods of high niose, signal lost for 24 seconds at 7h43m18 and for 30 seconds at 8h42m31|
|9911130935||98.98||X||1||HALCA signal late, signal lost for 24 seconds at 9h37m31|
|9911141052||99.32||Y||HALCA signal late, periods of high niose|
|9911150549||97.72||Y||2||two brief dropouts due to water tower, HALCA signal late, high noise for entire pass|
|9911151408||98.46||Y||1||HALCA signal late|
There was a total of 20 tracking passes in November of 1999 with 2 of these being an unattended tracking passes. Of the 18 attended tracking passes 16 collected the maximum amount of valid data possible. 1 of 2 unattended tracking passes collected the maximum amount of good data. 9 tracking passes lost data due to the link from HALCA not showing up on time. 4 tracking passes lost data due to twenty or thirty second dropouts of the HALCA signal.
18 tracking passes were observed ``remotely'' from the Jansky Operations Center.
16 of 18 attended passes had max good data.
Operators helped increase amount of good data in 2 of 2 ``bad'' attended passes.
1 of 2 unattended passes had max good data.
Operators would have increased amount of good data in 1 of 1 ``bad'' unattended passes.
A total of 1.86 tracking passes lost data due to hardware problems.
A total of 0.10 tracking passes lost data due to the signal from HALCA being late.
16 of 18 tracking passes observed from the JOC were succesfull.